23 Liter Class B white autoclave with USB and integrated printer // Clearance sale

2026-03-02T16:44:16+01:00

Vendor: Icanclave Autoclaves
Type: Autoclaves and accessories
Price: 2399.90

23 Liter Class B white autoclave with USB and integrated printer // Clearance sale

Refurbished by manufacturer

Technical specifications
- Voltage: 220-240 VAC; 50/60Hz; 10A
- External dimensions of the device: W x H x L: 490 x 455 x 690 mm
- Tray dimensions: 380 x 192 mm
- Circuit breaker: F16A / 400V
- Rated power: 1500 VA
- 1750W
- Sterilization temperature: 121°C/134°C
- Distilled water tank capacity: 2.5 L
- Operating temperature: 5°C/40
- Relative humidity: Max. 80%, non-condensing
- Net weight: 53 kg
- Maximum sound level: <70 dB
- Atmospheric pressure of 76 kPa / 106 kPa
- 338 Cycles
- Warranty: 3 months

Security
- It is essential that the water quality is below 20 microsiemens for the autoclave to start the cycles.
- The cycle does not start if the door is not properly closed.
- Automated safety with double hydraulic locking device, prevents the door from opening while the chamber is under pressure.
- Safety valves prevent pressure generator and camera.
- A safety thermostat prevents overheating in the chamber and generator
- An automatic safety lock prevents accidental opening of the door during the sterilization process protector automatically cuts off the electrical energy if somewhere has a short circuit or the energy supplier is not very stable.
- Failure alarm indicates a failure cycle and detailed error code.
- Door alarm indicates that the door is not secured.
- Low water alarm indicates when the water tank is empty or too full.

Icanclave autoclave: maintenance and error message guide

USER INSTRUCTIONS

Components
- Distilled water tank
- LCD screen
- Control panel
- Main switch
- Hose for draining distilled and used water
- Key to tighten the door
- USB port
- Internal printer (allows you to print sterilization cycle reports on thermal paper after the end of the cycle.)
- Safety valve
- Ventilation grille
- Locking system
- Power cable
- Label

Accessories included
- Sterilizer
-USB
- 4 instrument trays
- Tray support
- Handle for extracting the trays
- Door adjustment tool
- 2 drain pipes
- Tip for the door
- Instruction manual

Regulations
- 93/42 / CEE The medical care directive.
- 97/23/EEC pressure equipment directive.
- EN 13060 Small steam sterilizers.
- EN 61326, EN 61000 EMC.
- EN 61010-1, UL 61010-1 safety requirements.
- EN 61010-2-040, IEC 61010-2-040 Particular requirements for steam sterilizers used for the treatment of medical materials.

Autoclave process step by step
1. Open the door, take the tools and accessories from the sterilization chamber, remove the packaging and clean them.

2. Plug the power cord into an appropriate power source.

3. Plug in the printer.

4. Switching on: the main switch is located on the front, bottom right of the device. Once turned on, the LCD screen will light up and the basic operating symbols of the sterilizer, such as door position, water level, sterilization program, date, time will light up.

5. Remove the top cover then fill the reservoir with distilled water. If a beep sounds, this means that you have exceeded the allowable water level in the tank. Empty excess water until the alarm goes off.

6. Preparation of the equipment to be sterilized: place the instruments on the different trays, and separate them according to their materials. If the instruments are not stainless steel, use sterilization bags. These must not be in contact. Check that all the instruments are properly placed, so that water does not remain in the hollow parts. Do not overload instrument trays or place them in direct contact with the walls of the sterilization chamber. Do not use paper clips or needles to avoid damage or malfunction of the autoclave.

7. Select the sterilization program.

8. Start the sterilization cycle.
After selecting the sterilization program, place the instruments on the trays in the chamber using the plastic tweezers provided.
Once the instruments are securely placed inside, close the door. Caution: make sure to close the door securely, otherwise the sterilizer will not work.

9. Start the sterilization program.
Press the power button, the autoclave will start working for about 30-75 minutes.

10. End of the sterilization cycle.
Once the sterilization cycle is completed, the printer will automatically print the complete cycle report.
Once the pressure has dropped to 0, you can open the door and remove the instruments. The autoclave will notify you by sound once the sterilization cycle is completed.
Caution: Use the plastic tweezers to remove the instruments.

IMPORTANT: Do not reuse used water after emptying the tank.

It is important to keep the packaging box in case of after-sales service.

Program

Temp.

/Pressure

Sterilization time

Total time (min)

8D Pro | 12D Pro | 18D Pro | 23D Pro
QUICK 134 | 134℃/210Kpa | 3.5 mins | 10-20 | 11-20 | 12-30 | 15-30
WRAPPED 134 | 134℃/210Kpa | 4 mins | 25-40 | 28-45 | 30-49 | 38-57
WRAPPED 121 | 121℃/110Kpa | 20 mins | 36-46 | 37-48 | 38-54 | 46-62
PRION 134 | 134℃/210Kpa | 18 mins | 35-46 | 36-48 | 37-52 | 45-60
B&D TEST | 134 | 3.5 mins | 10-20 | 15-25 | 15-30 | 28-35
HELIX TEST | 134 | 3.5 mins | 10-20 | 15-25 | 15-30 | 28-35
VACUUM TEST | – | – | 17-20 | 18-21 | 19-22 | 20-25

* Cycle time is based on different loads and with/without preheat.

* Sterilization time and drying time can be adjusted.

Clearance condition: This product cannot be returned or exchanged.